Combinational External Fixator (C type)
1.It is made up of relatively independent components and ports.
2.According to clinical need, screw wires with different diameters can be used to carry out multipoint needle threading, multidirectional needle threading and crossover needle threading.
3.The functional part can be used either alone or in combination. It has not only functions of fixing, pulling, compressing, but also different kinds of connections for joints' activities.
4.It fits for regular intensity, which corresponds with the demands of biomechanics for bones' functional reconstruction.
5.Indications: limbs fracture, articular fracture, and pelvis fracture. Compressing fixation for bone disconnection and amputation, bone's extending for bone defect.
Method of Use
For the need for reduction of the fracture or dislocation, recommend the use of this method as a standard technique.
1. In each of the major fracture block is screwed into the 2 bone traction pin. According to the fracture, soft tissue conditions and local anatomy to select the entry pin position. If the bone traction pin distance more far, just bracket. The more stable.
2. Each of the major fracture block 2 bone traction pin with a fixed rod connection. Can use the spring fixing type pipe needle clamp to connect. Note that in the fracture zone have sufficient length fixing rod.
3. At the fracture site with third pole and 2 pole connecting pipe clip of wo rod end connection.
4. Then will connect bone traction pin support rod as the handle, for fracture reduction. In a radiographic image enhancement apparatus under the monitoring of reduction is confirmed, then maintaining reduction and screw down one by one tube clamp nut.
5. In order to facilitate the reset operation and reduce intraoperative X-ray irradiation: in each of the major fracture block two bone traction pin and the installation of a long rod, the rod can be used as a temporary lever, convenient reset, so as to make the operation more manipulative reduction, operating more effort, and operation the doctor’s hand can leave the X-ray radiation.
6. In the intraoperative or postoperative adjustment reset can loosen the hose clamp locking nut, according to the steps of adjusting the reduction of fracture. 24 hours later need to tighten all the locking nut.
1. Manufacturing & Trade combo for 12 years. We are professional in design, research, advanced technology, manufacture and sale etc.2. Quality assurance: CE; ISO13485; CHINA FAMOUS BRAND and over 10 Chinese National Patent Licenses, etc.3. Customization and OEM available.4. Export to many countries & areas, like India, Turkey, Middle East, Africa, Russia, US, UK, France, Brazil etc.5. Service & Support: 1 year warranty & online 24/7 service.6. Brand Assurance: PUSM Medical is one of the most famous orthopedic power tool brands all over the world.
Tel: +86 24 - 23519517
What'sApp: +86 1399 8124 587